Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||4 February 2010|
|PDF File Size:||4.50 Mb|
|ePub File Size:||5.10 Mb|
|Price:||Free* [*Free Regsitration Required]|
Guess you never know until you try! In the code above, MHz is an arbitrary frequency. Good luck with it I am able to communicate with the device jp4191a the GPIB interface, and am also able to retrieve information. Thanks for your help guys. When I gave it a try, though, it worked fine. Message 2 of 6.
I think the would probably take the old commands, but the ALSO has new commands. Accepted by topic author Matt C.
A RF IMPEDANCE ANL [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
If not, what would creating a driver for it entail? Message 6 of 6. I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen.
Has anyone done anything on interfacing with the A since Matt Mxnual last post? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. The forums emoticons took over for a bit there. It’s 84MB, so I hope you have a fast link.
I suspect that this difference might be causing my problems. Message 3 of 6.
Message 1 of 6. Message 4 of 6.
4191A RF IMPEDANCE ANL
Sorry, that last line of code there is supposed to read: You really need the manual. I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so mqnual anything I’ve ever seen before that I assumed it was referring to something different.
Message 5 of 6. However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at.
Agilent / HP 4191A RF Impedance Analyzer
Would you mind posting the code you used on the A? In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:. I have not worked with that instrument, but the Agilent web site has the operating and service manual here. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: Matt – you’re right, the commands for the A are completely different from the – probably because of its age.
Most Active Software Boards: IIRC, the sweep parameters were something like this: Should I be looking at doing it with LabView or something else?